Type 7334-1 Magnetic Emission Loop
RE01 and RE101 Magnetic Emission Tests
Type 7334-1 Loop Antenna has been
designed as a substitute for the AT-205/URM-6
antenna and is a replacement for Eaton Model
94607-1. It uses 36 turns of wire on a 5.25"
(13.3 cm) diameter form enclosed in an electrostatic
shield as described in paragraph 5.2.1 of
7334-1 is equipped with an epoxy-glass
base plate which serves as a spacer to enable the
user to place the loop at exactly 7 cm from the
face of the item under test as required by test
procedure RE01 of MIL-STD-462.
Connection to the loop is through a BNC connector which enables
coaxial cabling to the EMI receiver. The loop is supplied with a
correction factor graph showing the values in dB which must be added
to the reading of a 50 ohm EMI meter to obtain answers in either dB/μV/m
or dB/pT. The correction factor decreases as frequency increases
from 30 Hz up to approximately 15 KHz, where the factor levels off
and remains relatively constant up to 5 MHz.
The Type 7334-1 Loop Antenna is required by Test Method RE01 in
Parts 2 through 6 of MIL-STD-461C and RE101 of MIL-STD-461D.These
portions of the specification require magnetic field emission tests
of cables, equipments, systems and sub-systems installed in, or used
in, all phases of military vehicles, ships, submarines, aircraft
(including helicopters), spacecraft, or ground-based operations.
TEST METHODS RE01 and RE101
The Type 7334-1 Loop Antenna is positioned 7 cm from the face of the
equipment under test with the plane of the loop parallel to the
equipment face. The best position to begin with is opposite or near
a joint or seam.
The associated EMI meter is then scanned over the range 30 Hz to 100
KHz searching for emissions. At the frequencies where emissions are
found, the loop antenna is moved about the surface seeking the
strongest emission level.
When a strong signal is detected, the loop is oriented on its axis
for a maximum reading. This procedure is repeated for all surfaces
of the equipment under test. Although the specification is not clear
on the point, it appears to indicate that all six sides (including
the bottom) of an equipment must be tested in this manner.
When testing cables, the loop antenna is placed 7 cm from the cable
with the plane of the loop parallel to the cable. The non-metallic
base plate of the Type 7334-1 Loop Antenna provides a convenient
means for establishing the correct 7 cm distance.